Description & Technical characteristics
CT injector syringes for the radiopaque substances and physiological salt solution
Polymer container A capacity for a contrast agent, not less than 200 ml.
Polymer container B capacity for a physiological salt solution, not less than 200 ml.
Material — polyethylene terephthalate.
— 2 syringes, not less than 200 ml.
— polymer plunger with rubber cuff — 2 pcs.
— spiral connecting tubing, length not less than 150 cm, with T-connector — 1 pc.
— bag spike — 2 pcs.
Maximum design pressure, not less than 400 psi/2750 kPa.
Limiting rate of contrast agent injection — 10.0 ml/s.
Indicators in the form of ellipses on the surface of containers for visual control of the filling of syringes with contrast agent and physiological salt solution
Tube for collecting excess contrast agent, no more than 10 cm long, attached to the tubing
Shelf life from the date of sterilization, not less than 4 years.
Individual packing, sterile.
The supplier must provide a copy of the product registration certificate.